FEATURES: Use the intermational advanced technology. Clear imaging low leak radial, is only 1% of national standard Light touched button, controled with microcomputer, only press one button to select the exposue parameter, fast and accurately Developing in light room, image in one minute, the doctor can diagnose conveniently for maximum Pneumatic up-down seat, more convenient and more comfortable It is for the use of the digital imaging system of mouth Specification: Power supply:AC220V+10%, 50Hz, 1KVA Tube voltage:60kVp Tube current:8mA Focus size:1.5mm Total filtration:2.5mmAL Exposure time:0.2-4sec Leak radiation:Outside one meter. In order to operate safe, must be installed properly good ground. X-ray machine due to small thermal capacity is limited, should pay attention to intermittent operation is recommended.
The machine only during exposure, will produce X-rays, power-off state does not produce X-rays, as long as it does not produce radiation exposure is not under power on state. The machine is in the correct use, any unforeseen circumstances, please immediately cut off the power operator, discontinue use. After the aircraft maintenance standards, it can be used again. The machine factory, each X-ray device is equipped with the following specifications: 1.0 / 0.5mm aluminum each one. When disassembling the machine each time, first disassemble the nose and then split the other member.
After each use the machine, arm recovery highest point, the position of the head is placed equilibrium position. After each use of medical alcohol exposure in some patients with end beam tube disinfection, cleaning.
The weekly machine surface with a dry cloth to clean, inspect each connector contact. Note that the machine maintenance and inspection indicator that shows whether the knobs, switches, remote control is working properly; X-ray source assembly if possible oil leakage; when the X-ray source assemblies exposure, abnormal noise; mechanical portions of the fastening screws loosening the like.
The machine rated capacity of 1.1kW, please use the greater than 1.5kW power supply capacity or grid. If the regional grid voltage is not normal, we recommend supporting more than 1.5KW industrial regulator, to prevent damage to the machine voltage is not normal, reducing machine life. IF you want more, please contact me , i would like to give you some. STATEMENT(FDA, ARTG): The sale of this item may be subject to regulation by the U. Food and Drug Administration and state and local regulatory agencies.
The item "1 PIECE Wall Mounted style Digital Dental X-ray Machine" is in sale since Friday, April 19, 2019. This item is in the category "Business & Industrial\Healthcare, Lab & Dental\Medical & Lab Equipment, Devices\X-Ray Machines & Equipment\Other X-Ray Equipment". The seller is "jinrun-0" and is located in . This item can be shipped worldwide.