The radiation scope is at an angle of 24 degrees located within a distance of 1.2 meters ahead. This is an innovative Chinese design named the Green X-ray Machine. All the components of this product were originally imported to achieve an acceptable tolerance required for the digital controls and the machine. By doing this product perform ance is more stable and reliable. Using a Toshiba 0.3 x 0.3 micro-focus tube the image is clearer and more delicate.
This machine uses components from well established suppliers allowing us to deliver to our customers a product of high quality. There are films included for testing purpose and all the consumables can be replaced by other similar products. There are vacuum components and high-pressure parts inside of the machineso for your safety and the maintenance of the machineplease do not open the machine. Service and maintenance should only be performed by a trained professional.The tube head is equipped with excellent radiation-shield protection except the radial mouthand tests have shown that the radial index is zero. Non-film inspection and storage can be realized when corresponding to a sensor. Exposure time: 1-9 seconds optional. Storage conditions: 5-40 degrees Celsius, relative Humidity. Net weight: 5.5 kg.
Dimension: 47 x 42 x 19 cm. The Angle should be proper while taking a picture. The film should be put vertically behind the tooth to be taken a picture while the case is for the front teeth; and the film should be put transversely behind the tooth to be taken a picture while the case is for the back teeth.
The patient should use forefinger to press the film behind the tooth gently. Kodak film as the sent samples are strongly recommended.
Reference of the time of exposure. Operation of the Main Unit. Make sure the fittings on the encasement list are packaged within the box. Connect the six-core cable (of the Main Unit) to the joint of the ball head and tighten the screw; Install the blue ray tube to the foreside of the ball headthen tighten it.After the Main Unit and ball head are perfectly fit togetherplug in the international standard electrical sourcegrounding is strongly recommended. Turn on the Powerthe pilot lamp will be light up. During this timethe digital tube of the main unit will show the fore setting time. Now the equipment is in standby mode. Setting the time (Skip to next step if unneeded) Use the "SET""+"and" -" buttons to reset the time as follows: press SETthe pilot lamp above is illuminated the machine is now in the time setting mode? Then press "+" and" -" to set the time needed (time range is 1-9seconds); press set again when the time is adjusted the light lamp is off. The equipment is now in standby mode. Put the Tooth film plumb behind the tooth which is going to be taken picture and be as close as possible (the smooth side stick to the tooth). Keep the ball head plumb to the tooth projection position; Have the ball headtooth and tooth film steadily mutually plumb.
After positioninguse the ON/OFF on the main Unitthe remote controlleror the button on the touch head tube to take pictures. The three methods have the same effect.(Notice: gently press 0.5 sec to turn it on). After exposure to the tubepress the start button again (available in three ways) in case of an emergency to stop shooting and return to standby status. (At this point there was no output rayso the dental films can be used again). Turn the power of the main unit off when the work is finishedpress POWER making sure the pilot lamp is off. Don't keep the main unit long in standby mode.
Picture the tooth following the right operating procedure of the X-ray Unitthen take the film out from the mouth. Use a syringe to extract 4-5 mi of imaging liquid. Transfuse the liquid into the bagnip both the ends and the sides of the bag with thumbs and first fingers(knead the film).
Open the end of the bag about ten minutes after transfusing the liquid into the bag. Take the film output it into clear waterand rinse it. Don't wipe the surface of the film with the surface to prevent damage to the image. Observe and diagnose after the film becomes dry.The sale of this item may be subject to regulation by the U. Food and Drug Administration and. State and local regulatory agencies. If the item is subject to FDA regulation. If you have questions about legal obligations regarding sales of medical devices.
You should consult with the FDA's Center for Devices and Radiological Health. Thank you for your cooperation. We try our best to reply to your emails as soon as possible, however, due to high volume of daily incoming emails and time zone difference, we may not be able to reply your emails immediately. Please give us the opportunity to resolve any problem. We understand the concerns and frustrations you might have, and will try our best to resolve the issues.
We are expecting Good co-operation with you and hope to see you again. Feedback is very important to us.It is impossible to address issues if we do not know about them! Faction is our ultimate goal. We work hard to earn your trust and future business. A If you receive a defective item, you may return it or ask for a discount. Due to time zones, our reply to your questions may not be immediately.
If it's non-workdays, please kindly be patient and wait till Monday. The item "BLX-5 Dental X Ray Mobile Film Imaging Machine Digital Low-Dose System Portable" is in sale since Sunday, May 17, 2020. This item is in the category "Business & Industrial\Healthcare, Lab & Dental\Medical & Lab Equipment, Devices\X-Ray Machines & Equipment\Other X-Ray Equipment". The seller is "autoneighborstore" and is located in Rowland Heights, California.This item can be shipped to United States.