Focal Spot To Skin Distance. Reference of the time of exposure.
Operation of the Main Unit. Make sure the fittings on the encasement list are packaged within the box. Install the blue ray tube to the foreside of the ball head. The pilot lamp will be light up. The digital tube will show the fore setting time.Now the equipment is in standby mode. Setting the time (Skip to next step if unneeded) Use the "SET".
And" -" buttons to reset the time as follows: press SET. The pilot lamp above is illuminated. The machine is now in the time setting mode? Then press "+" and" -" to set the time needed (time range is 1-9seconds); press set again when the time is adjusted the light lamp is off. The equipment is now in standby mode.Put the Tooth film plumb behind the tooth which is going to be taken picture. And be as close as possible (the smooth side stick to the tooth). Keep the ball head plumb to the tooth projection position; Have the ball head. And tooth film steadily mutually plumb. Use the ON/OFF on the main Unit.
(Notice: gently press 0.5 sec to turn it on). After exposure to the tube.Press the start button again in case of an emergency to stop shooting and return to standby status. At this point there was no output ray. So the dental films can be used again. Please repeat steps 7 to 9. Turn the power off when the work is finished. Making sure the pilot lamp is off. Don't keep the machine long in standby mode. Make sure the angle of film. And ball head are properly aligned when taking a picture. Keep them steady until the last step of picturing; making sure that there is no change in any of their positions. Remember to turn the power 0ff when work is done. Charge the battery if the machine cannot work and the red lamp lignt. After turning the power on. Wait one minute before taking pictures. This allows the ray to provide a steady output. The equipment switches into protection mode automatically when the voltage is incorrect. It will not be able to carry out the normal work of filming. The ON/OFF button could stop the objection and then the machine goes back to the preparation mode. Use high quality tooth film and developing liquid to make clear pictures. Handle the ball head gently while in use. So as not to damage the delicate component. We usually set the time between 3-8 seconds when taking pictures. But the actual time of ray output is 0.2-0.5 seconds. Preventing the machine from taking regular pictures. Keep the tooth film and the liquid in a proper place and use them within the shelf-life period.
Development of films should be performed at temperatures ranging between 23-25 degrees Celsius to ensure the image quality. Other requests accordance with the scope of the technical parameters.
Please read the manual carefully before using. The sale of this item may be subject to regulation by the U. Food and Drug Administration and state and local regulatory agencies. The item "Digital Dental PORTABLE Green X-ray Machine Hight Frequency 30 KHz 24V UPS" is in sale since Thursday, March 8, 2018. This item is in the category "Business & Industrial\Healthcare, Lab & Dental\Medical & Lab Equipment, Devices\X-Ray Machines & Equipment\Other X-Ray Equipment".The seller is "dentalbaistra" and is located in Fullerton, California. This item can be shipped to United States.