This model combines the advantages of similar products found at home and abroad; it eliminates the shortcomings of the on-frequency X-ray machine (high current intensity and excess amounts of scrap x-rays). Using a Toshiba 0.3 micro-focus tube. The tube voltage frequency of this machine is 30 KHzand the tube current is 1 MA.
The radiation scope is at an angle of 24 degrees located within a distance of 1.2 meters ahead. This is an innovative Chinese design named the Green X-ray Machine. Focal Spot To Skin Distance. Reference of the time of exposure. Operation of the Main Unit.
Make sure the fittings on the encasement list are packaged within the box. Turn on the Powerthe pilot lamp will be light up. During this timethe digital tube will show the fore setting time. Now the equipment is in standby mode. Setting the time (Skip to next step if unneeded) "+"and" -" buttons to reset the time as follows: press SETthe pilot lamp above is illuminated the machine is now in the time setting modethen press "+" and" -" to set the time needed (time range is 0.1-1.8seconds); The equipment is now in standby mode.Put the Tooth film plumb behind the tooth which is going to be taken picture and be as close as possible (the smooth side stick to the tooth). Keep the ball head plumb to the tooth projection position; Have the ball headtooth and tooth film steadily mutually plumb. After positioninguse the ON/OFF on the main Unit to take pictures. (Notice: gently press 0.5 sec to turn it on).
Make sure the angle of filmtoothand ball head are properly aligned when taking a picture. Keep them steady until the last step of picturing; making sure that there is no change in any of their positions. Remember to turn the power 0ff when work is done. Charge the battery if the machine cannot work and the red lamp lignt.After turning the power onwait one minute before taking picturesthis allows the ray to provide a steady output. The equipment switches into protection mode automatically when the voltage is incorrect. If this occurs it will not be able to carry out the normal work of filming. When taking picturesthe ON/OFF button could stop the objection and then the machine goes back to the preparation mode. Use high quality tooth film and developing liquid to make clear pictures. Handle the ball head gently while in useso as not to damage the delicate component. Keep the tooth film and the liquid in a proper place and use them within the shelf-life period.
Development of films should be performed at temperatures ranging between 23-25 degrees Celsius to ensure the image quality. Other requests accordance with the scope of the technical parameters. Please read the manual carefully before using. FDA Statement:The sale of this item may be subject to regulation by the U.
Food and Drug Administration and state and local regulatory agencies. We commited ourselfs to your 100% satisfaction both with our goods and services.
Please CONTACT me immediately before leave us neutral feedback or negtive feedback, We will try our best to resolve any problem for you untill you are satisfied. 20-30 days to UK, Germany, French, Australia, Canada, etc, Over 30 days to other countries. Our guarantee does not cover accidental damage or wear of product caused by customer. Thank you very much for your understanding! We're one of the largest online dental suppliers based on many qualified manufacturers in china.
Looking forward to our friendly & win-win cooperation! The item "Portable Dental X-Ray Machine Unit Green X-ray Machine" is in sale since Tuesday, June 30, 2020. This item is in the category "Business & Industrial\Healthcare, Lab & Dental\Medical & Lab Equipment, Devices\X-Ray Machines & Equipment\X-Ray Machines/Generators".The seller is "dentalmed" and is located in Fullerton, California. This item can be shipped to United States.