The radiation scope is at an angle of 24 degrees located within a distance of 1.2 meters ahead. This is an innovative Chinese design named the Green X-ray Machine. High pressure generator 30kHz DC. Focal Spot To Skin Distance >20CM. Focus Spot Size 0.3mm(TOSHIBA).Reference of the time of exposure. Central & Lateral 0.8-1.2 0.8-1.0. Cuspid & 1st Bicuspid 1.2-1.4 1.0- 1.2. Upper 56 1.6-1.8 1.4-1.6. Upper 78 3.0-3.5 2.8-3.3. Lower 56 1.8-2.0.
Lower 78 2.0-2.5 1.8-2.3. Operation of the Main Unit. Make sure the fittings on the encasement list are packaged within the box. Install the blue ray tube to the foreside of the ball headthen tighten it. Turn on the Powerthe pilot lamp will be light up.
During this timethe digital tube will show the fore setting time. Now the equipment is in standby mode.
Setting the time (Skip to next step if unneeded) Use the "SET""+"and" -" buttons to reset the time as follows: press SETthe pilot lamp above is illuminated the machine is now in the time setting modethen press "+" and" -" to set the time needed (time range is 1-9seconds); press set again when the time is adjusted the light lamp is off. The equipment is now in standby mode.
Put the Tooth film plumb behind the tooth which is going to be taken picture and be as close as possible (the smooth side stick to the tooth). Keep the ball head plumb to the tooth projection position; Have the ball headtooth and tooth film steadily mutually plumb. After positioninguse the ON/OFF on the main Unit to take pictures. (Notice: gently press 0.5 sec to turn it on). After exposure to the tubepress the start button again in case of an emergency to stop shooting and return to standby status.
(At this point there was no output rayso the dental films can be used again). Turn the power off when the work is finishedpress POWER making sure the pilot lamp is off. Don't keep the machine long in standby mode.
Make sure the angle of filmtoothand ball head are properly aligned when taking a picture. Keep them steady until the last step of picturing; making sure that there is no change in any of their positions. Remember to turn the power 0ff when work is done. Change the battery if the machine cannot work and the red lamp lignt. After turning the power onwait one minute before taking picturesthis allows the ray to provide a steady output.
The equipment switches into protection mode automatically when the voltage is incorrect. If this occurs it will not be able to carry out the normal work of filming. When taking picturesthe ON/OFF button could stop the objection and then the machine goes back to the preparation mode. Use high quality tooth film and developing liquid to make clear pictures. Handle the ball head gently while in useso as not to damage the delicate component.
Do not pull hard on the cable which connects the ball head to the main unit. The cable connecting the ball head to the main unit cannot bear the weight of the ball head, so do not raise the ball head by holding the cable, it will cause damage.
We usually set the time between 3-8 seconds when taking picturesbut the actual time of ray output is 0.2-0.5 seconds. Keep the tooth film and the liquid in a proper place and use them within the shelf-life period. Development of films should be performed at temperatures ranging between 23-25 degrees Celsius to ensure the image quality. Other requests accordance with the scope of the technical parameters. Please read the manual carefully before using.About X-ray machine battery problem, lithium battery inside needs to keep active and can't be stored for a long time without usage, which can lead to power failure. If it is not used for a long time, they are advised to discharge properly, (turn on the screen or expose several times), then charge about one hour. The sale of this item may be subject to regulation by the U.
Food and Drug Administration and state and local regulatory agencies. EUB (under 2KG): Only for USA, Canada, and Australia.China Post (under 2 KG): 25-55 days. Few states need to pay remote surcharge. Don't worry, I will tell you before you pay if it's need. Please make sure it's correct.
Your SATISFACTION is our ultimate goal. We work hard to earn your trust and future business. A Please check the package first before you sign for the package. B If you have a defective item, you can return it or ask for a discount.
Or replace item for you. We are factory founded dental equipment Wholesaler and Retailer.If you have any questions about our service or products. You will be replied within 24 hours (48 hours in the weekend). The item "Dental Digital X-Ray Imaging System Mobile X Ray Machine Unit BLX-8 Plus" is in sale since Friday, October 23, 2020.
This item is in the category "Business & Industrial\Healthcare, Lab & Dental\Medical & Lab Equipment, Devices\X-Ray Machines & Equipment\Other X-Ray Equipment". The seller is "dentaltop1" and is located in Shenzhen. This item can be shipped worldwide.