This is an innovative Chinese design named the Green X-ray Machine. Focal Spot To Skin Distance. Reference of the time of exposure. Operation of the Main Unit.
Make sure the fittings on the encasement list are packaged within the box. Install the blue ray tube to the foreside of the ball headthen tighten it. Turn on the Powerthe pilot lamp will be light up. During this timethe digital tube will show the fore setting time. Now the equipment is in standby mode.Setting the time (Skip to next step if unneeded) Use the "SET""+"and" -" buttons to reset the time as follows: press SETthe pilot lamp above is illuminated the machine is now in the time setting modethen press "+" and" -" to set the time needed (time range is 1-9seconds); press set again when the time is adjusted the light lamp is off. The equipment is now in standby mode. Put the Tooth film plumb behind the tooth which is going to be taken picture and be as close as possible (the smooth side stick to the tooth). Keep the ball head plumb to the tooth projection position; Have the ball headtooth and tooth film steadily mutually plumb. After positioninguse the ON/OFF on the main Unit to take pictures.
(Notice: gently press 0.5 sec to turn it on). After exposure to the tubepress the start button again in case of an emergency to stop shooting and return to standby status. (At this point there was no output rayso the dental films can be used again).
Turn the power off when the work is finishedpress POWER making sure the pilot lamp is off. Don't keep the machine long in standby mode. Make sure the angle of filmtoothand ball head are properly aligned when taking a picture. Keep them steady until the last step of picturing; making sure that there is no change in any of their positions. Remember to turn the power 0ff when work is done.Charge the battery if the machine cannot work and the red lamp lignt. After turning the power onwait one minute before taking picturesthis allows the ray to provide a steady output.
The equipment switches into protection mode automatically when the voltage is incorrect. If this occurs it will not be able to carry out the normal work of filming. When taking picturesthe ON/OFF button could stop the objection and then the machine goes back to the preparation mode. Use high quality tooth film and developing liquid to make clear pictures.
Handle the ball head gently while in useso as not to damage the delicate component. Do not pull hard on the cable which connects the ball head to the main unit. The cable connecting the ball head to the main unit cannot bear the weight of the ball head, so do not raise the ball head by holding the cable, it will cause damage.
We usually set the time between 3-8 seconds when taking picturesbut the actual time of ray output is 0.2-0.5 seconds. Keep the tooth film and the liquid in a proper place and use them within the shelf-life period. Development of films should be performed at temperatures ranging between 23-25 degrees Celsius to ensure the image quality. Other requests accordance with the scope of the technical parameters. Please read the manual carefully before using.About X-ray machine battery problem, lithium battery inside needs to keep active and can't be stored for a long time without usage, which can lead to power failure. If it is not used for a long time, they are advised to discharge properly, (turn on the screen or expose several times), then charge about one hour.
Statement:The sale of this item may be subject to regulation by the U. Food and Drug Administration and state and local regulatory agencies.
Surgical Handpiece Straight Nose Cone 1:1 With External Pipe NEW -Gift. Air pressure:245-392KPa Water consumption:50ML/min Air consumption:1.5L/min Rotation speed:14,000-20,000r/min Noise70dB Sterilization temperature:135 Metal rotary sleeve, can resist the high temperature. Packing List: Handpiece1 Operation manual1.Our guarantee does not cover accidental damage or wear of product caused by customer. Thank you very much for your understanding! You feedback is VERY important to us. We understand the concerns and frustrations you might have, and will try our best to resolve the isuues. Serivce is our #1 priority at BAISTRA! We offer the best items at the best prices. We strive to exceed all your expectations and put a smile on your face. We stand behind our products and pride ourselves on OUTSTANDING customer service.
The sale of this item may be subject to regulation by the U. The item "Dental Portable X-Ray Machine Mobile Digital Unit System +11 Surgical Handpiece" is in sale since Saturday, February 23, 2019. This item is in the category "Business & Industrial\Healthcare, Lab & Dental\Medical & Lab Equipment, Devices\X-Ray Machines & Equipment\Other X-Ray Equipment". The seller is "dentalbravo" and is located in Fullerton, California.
This item can be shipped to United States.