This model combines the advantages of similar products found at home and abroad; it eliminates the shortcomings of the on-frequency X-ray machine (high current intensity and excess amounts of scrap x-rays). Using a Toshiba 0.3 micro-focus tube. The tube voltage frequency of this machine is 30 KHz.And the tube current is 1 MA. The radiation scope is at an angle of 24 degrees. Located within a distance of 1.2 meters ahead.
This is an innovative Chinese design named the Green X-ray Machine. Warning: After turning on the machine, wait for 1 minute, the voltage can be stabilized before normal use. After each photo, you have to wait 30 seconds before you can take a normal photo. Focal Spot To Skin Distance.Reference of the time of exposure. Operation of the Main Unit. Make sure the fittings on the encasement list are packaged within the box.
The pilot lamp will be light up. Will show the fore setting time. Now the equipment is in standby mode. Setting the time (Skip to next step if unneeded). Then press "+" and" -" to set the time needed (time range is 0.1-1.8seconds); The equipment is now in standby mode.
Put the Tooth film plumb behind the tooth which is going to be taken picture. And be as close as possible (the smooth side stick to the tooth). Keep the ball head plumb to the tooth projection position; Have the ball head. And tooth film steadily mutually plumb.
Use the ON/OFF on the main Unit. (Notice: gently press 0.5 sec to turn it on). Make sure the angle of film. And ball head are properly aligned when taking a picture.
Keep them steady until the last step of picturing; making sure that there is no change in any of their positions. Remember to turn the power 0ff when work is done. Charge the battery if the machine cannot work and the red lamp lignt.
After turning the power on. Wait one minute before taking pictures. This allows the ray to provide a steady output.The equipment switches into protection mode automatically when the voltage is incorrect. It will not be able to carry out the normal work of filming. The ON/OFF button could stop the objection and then the machine goes back to the preparation mode. Use high quality tooth film and developing liquid to make clear pictures. Handle the ball head gently while in use. So as not to damage the delicate component. Preventing the machine from taking regular pictures. Keep the tooth film and the liquid in a proper place and use them within the shelf-life period. Development of films should be performed at temperatures ranging between 23-25 degrees Celsius to ensure the image quality. Other requests accordance with the scope of the technical parameters. Please read the manual carefully before using. The sale of this item may be subject to regulation by the U. Food and Drug Administration and state and local regulatory agencies.
System, X-Ray, Mobile K200976 SR-8230, SR-8230S Portable X-Ray Unit 892.172 IZL 4/13/2020 6/10/2020 Substantially Equivalent (SESE) Radiology Radiology Summary Traditional No No. The item "Portable Dental Digital X-Ray Imaging System Mobile X-Ray Machine BLX-5(8PLUS)" is in sale since Wednesday, January 10, 2018. This item is in the category "Business & Industrial\Healthcare, Lab & Dental\Handpieces & Instruments\Dental Instruments". The seller is "deerbell" and is located in Fullerton, California.This item can be shipped to United States.