Non-film inspection and storage can be realized when corresponding to a sensor Specifications. Exposure time: 1-9 seconds optional. Storage conditions: 5-40 degrees Celsius, relative Humidity.
Net weight: 5.5 kg. Dimension: 47 x 42 x 19 cm Content. Dental Film Position x 1. Make sure the fittings on the encasement list are packaged within the box.
Connect the six-core cable (of the Main Unit) to the joint of the ball head and tighten the screw. After the Main Unit and ball head are perfectly fit together, plug in the international standard. Electrical source, grounding is strongly recommended. Turn on the Power; the pilot lamp will be light up. During this time, the digital tube of the main unit.
Will show the fore setting time. Now the equipment is in standby mode. Setting the time (Skip to next step if unneeded) Use the "SET", "+", and" -" buttons to reset the time.
As follows: press SET, the pilot lamp above is illuminated, the machine is now in the time setting. Modethen press "+" and" -" to set the time needed (time range is 0.5-9seconds); press set again when. The time is adjusted the light lamp is off. The equipment is now in standby mode. Put the Tooth film plumb behind the tooth which is going to be taken picture, and be as close as possible (the smooth side stick to the tooth).Keep the ball head plumb to the tooth projection position; Have the ball head, tooth, and tooth film steadily mutually plumb. After positioning, use theEXP on the main Unit, the remote controller, or the button on the touch head tube to take pictures. The three methods have the same effect. After exposure to the tube, release the start button again (available in three ways) in case of an emergency to stop shooting and return to standby status. (At this point there was no output ray, so the dental films can be used again). Turn the power of the main unit off when the work is finished, press POWER, making sure the pilot.
Don't keep the main unit long in standby mode. Make sure the angle of film, tooth, and b. All head are properly aligned when taking a picture.Keep them steady until the last step of picturing. Making sure that there is no change in any of their positions. Remember to turn the power offwhen work is done. Change the battery if the remote controller loses its functioning or if the remote's distance is shorted. After turning the power on, wait one minute before taking pictures, this allows the ray to provide a. The equipment switches into protection mode automatically when the voltage is incorrect. Occurs, it will not be able to carry out the normal work of filming. When taking pictures, the EXP button could stop the objection and then the machine goes back to the.
Use high quality tooth film and developing liquid to make clear pictures. Handle the ball head gently while in use, so as not to damage the delicate component.
Do not pull hard on the cable which connects the ball head to the main unit. The cable connecting the ball head to the main unit cannot bear the weight of the ball head, so do not raise the ball head by holding the cable, it will cause damage. We usually set the time between 3-8 seconds when taking pictures, but the actual time of ray output. Is 0.2-0.5 seconds. Keep the tooth film and the liquid in a proper place and use them within the shelf-life period.Development of films should be performed at temperatures ranging between 23-25 degrees Celsius to ensure the image quality. Other requests accordance with the scope of the technical parameters. Please read the manual carefully before using. The sale of this item may be subject to regulation by the U. About X-ray machine battery problem, lithium battery inside needs to keep active and cant be stored for a long time without usage, which can lead to power failure.
If it is not used for a long time, they are advised to discharge properly, (turn on the screen or expose several times), then charge about one hour. We commited ourselfs to your 100% satisfaction both with our goods and services. Please message me immediately before leave us neutral feedback or negtive feedback, We will try our best to resolve any problem for you untill you are satisfied. If the item you received is damaged or defective, Please Leave message immidately.
Our guarantee does not cover accidental damage or wear of product caused by customer. Thank you very much for your understanding! We're one of the largest online dental suppliers based on many qualified manufacturers in china.
If you Leave message immediately after placing the order, we will make every effort to assist you with your change request. Any help, need or questions, please do not hesitate to Leave message.Looking forward to our friendly & win-win cooperation! The item "UPS Portable Dental Digital X-Ray Unit Imaging Mobile Machine" is in sale since Thursday, December 27, 2018. This item is in the category "Business & Industrial\Healthcare, Lab & Dental\Medical & Lab Equipment, Devices\X-Ray Machines & Equipment\Other X-Ray Equipment".
The seller is "dentistryshop" and is located in Fullerton, California. This item can be shipped to United States.